The PiezoDrive AP Series displacement actuators utilize an advanced flexural mechanism and high performance piezoelectric stack actuators to provide an exceptionally large range of motion, fast response, and sub-nanometer resolution. The advanced dual-hinge flexure mechanism significantly outperforms competing devices due to its significantly higher in-plane and out-of-plane stiffness, greater mechanical efficiency, and higher resonance frequency. The dual-hinge flexure design also results in extremely compact dimensions.

Applications include: Nanopositioning, Biomedical, Microscopy, Precision machining, Vibration control, High-speed valves, and Optics. A range of standard devices are available for immediate delivery, however, custom dimensions and travel ranges can be produced with a four-week lead time.

Displacement | >120 um |

Voltage | -15V to +150V |

Unipolar Disp. | >110 um |

Resonance | 900 Hz |

Force | 11 N |

Stiffness | 0.10 N/um |

Capacitance | 400 nF |

Mass | 5 g |

3D Model | AP120.igs |

Displacement | >350 um |

Voltage | -15V to +150V |

Unipolar Disp. | >320 um |

Resonance | 400 Hz |

Force | 6 N |

Stiffness | 0.019 N/um |

Capacitance | 900 nF |

Mass | 7 g |

Displacement | >340 um |

Voltage | -15V to +150V |

Unipolar Disp. | >310 um |

Resonance | 460 Hz |

Force | 16 N |

Stiffness | 0.052 N/um |

Capacitance | 780 nF |

Mass | 8 g |

Displacement | >350 um |

Voltage | -15V to +150V |

Unipolar Disp. | >320 um |

Resonance | 480 Hz |

Force | 18 N |

Stiffness | 0.057 N/um |

Capacitance | 900 nF |

Mass | 9 g |

3D Model | AP350.igs |

Displacement | >830 um |

Voltage | -15V to +150V |

Unipolar Disp. | >750 um |

Resonance | 230 Hz |

Force | 90 N |

Stiffness | 0.12 N/um |

Capacitance | 8.3 uF |

Mass | 60 g |

3D Model | AP830.igs |

Each actuator is individually tested for displacement range, blocking force, and resonance frequency. A test report is provided with each delivery.

The AP actuators can be mounted in a fixed-free or free-free configuration using the threaded mounting holes. When a voltage is applied, a proportional expansion is developed in the vertical direction, as illustrated below.

The maximum tension force is 10% of the blocking (compressive) force.

The range is specified for an applied voltage of -15V to +150V. The unipolar range is from 0V to +150V.

When an amplified actuator is driving a spring, the range is reduced by the factor $$\frac{k_a}{k_a+k_L}$$ where \(k_a\) is the actuator stiffness and \(k_L\) is the load stiffness.

The actuator capacitance is the small-signal capacitance measured at room temperature. Due to hysteresis, the effective capacitance increases with applied voltage. When operating at full range, the effective capacitance is approximately twice the small-signal capacitance. The capacitance also increases with temperature. A temperature increase of approximately 50 degrees C will double the effective capacitance.

The required current is \( I = C~ dV/dt \) where \( I \) is the current, \( C \) is the effective capacitance, and \( dV/dt \) is the voltage rate of change. For a sine-wave, the required peak current is equal to: $$ I_p = 2 \pi f V_{p-p} $$ where \( V_{p-p} \) is the peak-to-peak voltage. For a triangle wave, the required peak current is equal to: $$ I_p = 2 C f V_{p-p} $$

Amplifier | Application |

PDm200B | General purpose, low cost |

MX200 | High Current, Low noise, low cost |

PDu150 | Ultra-low noise |

PD200 | High speed, low noise |

PX200 | High current, low noise |

The AP Actuators and wiring insulation meet the outgassing requirements for NASA SP-R-0022A.

Custom dimensions and travel ranges can be produced with a four-week lead time.

Copyright © 2010 *Piezo*Drive

part of Newcastle Innovation Ltd

site by: unconundrum.net